The flow rate calculator for EMED Technologies flow control devices, is a tool designed to assist clinicians in the selection of infusion products and their appropriate settings to ensure a safe and comfortable infusion experience. Follow these easy steps to use and understand the online flow rate calculator.
Select your country or region
Select infusion drug & enter dosage amount in milliliters
NOTE: Ensure that you are familiar with the prescribing information for your drug applicable to your country.
Some drugs also require information about the patient weight category. Select the appropriate weight category if the dropdown box appears.
Select the type of infusion
For people performing infusions for the first time, select Initial Infusion. For people performing regular or maintenance infusions, select the Maintenance Infusion option.
Select your infusion pump
Select an appropriate needle set
Number of needles
Specific needle set code
Select the flow control device
For fixed rate control, select Infuset, and for variable rate control, select either VersaRate or VersaRate Plus.
For variable flow control: Select flow controller position setting
For fixed flow control: Select specific Infuset code
Understanding Your Infusion Results
Total flow rate
This value represents the average total flow rate of your drug which is delivered to all of your infusion sites combined during the full duration of the infusion. It is shown in milliliters per hour (mL/hr).
Flow rate per site
This value represents the average flow rate of your drug delivered to each of your infusion sites individually during the full duration of the infusion. It is shown in milliliters per hour (mL/hr).
Dosage volume per site
The dosage volume per site is the volume of drug (in milliliters) that will be supplied through each needle to each of your infusion sites.
The infusion time is the expected duration that your infusion therapy session will take.
Notes and Warning Messages
The flow rate calculator has been designed to warn the user when the combination of selected infusion products may exceed the prescribing limits for the chosen drug. Depending on the type of infusion selected, these warnings may appear when exceeding either the total flow rate for all needle sites combined, the flow rate limit per needle site, and the dosage volume allowed per needle site. Warning messages will appear in the Infusion Results section in the colors described below.
Warnings and cautions regarding flow rate and volume limitations are based on recommendations made by the drug manufacturer of the selected drug and may be subject to change. All reasonable attempts are made to ensure that this calculator contains the most recent drug related information, but it is the user's responsibility to ensure familiarity with the relevant drug and its most recent instructions for use for their particular country before using this calculator.
Warnings in orange
Warning messages that appear in orange indicate that the chosen combination of infusion products is expected to exceed the prescribed limit of that particular drug for patients performing initial infusions.
Warnings in red
Warning messages that appear in red indicate that the chosen combination of infusion products is expected to exceed the prescribed limit of that particular drug for patients performing regular or maintenance infusions.
NOTE: The flow rates shown in this tool shall be used as a guide/reference only to illustrate the capabilities of the flow control/regulator sets when used to infuse the selected drug in combination with the selected parameters. These are a representation of expected values using the SCIg60 Infusion System based on testing at room temperature between 20°C - 25°C (68°F - 77°F), and may vary based on different needle lengths, fluctuations in temperature (which significantly affects the viscosity of the selected drug), large height differences between the infusion pump and infusion site, and patient anatomy. Flow rates shown represent an average expected flow rate, and may in some cases, for the VersaRate and VersaRate Plus, be lower or higher by up to ±40% at position settings 1-2, up to ±20% at position settings 3-4 and up to ±15% at higher position settings. This is a typical range of variability for these types of adjustable flow control devices. For the Infuset fixed rate controller, average flow rates may be higher or lower by up to ±15% due to system tolerance and the sources of variation already mentioned. For more information, please contact EMED Technologies.